|Weight:||About270g(Include Battery)||Max Display:||3999 Or 9999|
|Max Expand Of The Pincers:||30mm||Product Model:||Clamp Meter Bm819 Digital Multimeter|
Bm819 Digital Clamp Tester,
100uF Digital Clamp Tester,
40K multimeter digital clamp
Clamp Meter Bm819 Digital Multimeter
BM819 OPERATOR'S MANUAL
Welcome to use this product.
The digital clamp meter adopts high-performance MCU processor,
it is great value as has high reliability,security, automatically setting
range funcition,hand-held clamp meter etc.. the product has a large
digital display, full range overload protection, data-hold function,
undervoltage indication, auto shut off function,it has TRUE RMS
measurement function which can accurately measure frequency
voltage, non-sinusoidal voltage, it also has temperature measurement
function which can accurately measured -50~1000 ºC, the instrument
is suitable for frequency-converting power supply, air conditioning,
refrigeration equipment such as refrigerators, motor performance test,
It is more excellent for a new generation of practical electrical
measuring instrument .
This product is ideal instrument for test, maintenance and repais in
college, smelting, communications, manufacturing, petroleum, electric
power, national defense,electronic, electric power equipment .
The meter in structure complies with the safety requirements of
ICE61010-1 CAT 600V. Read the instruction carefully as follows
before you use the meter:
1. When measuring voltage, AC or DC voltage should not be more than
the voltage (DC / AC 600V RMS) of the meter.
2. Voltage of less than 36V is safety voltage. When the voltage is more
than DC 51V or AC36V, the leads should be checked. The test lead
should be connected correctly and their insulating property should be
under excellent status against electric shock.
3. When change of functional measuring range, the test lead should be
away from test point.
4.It is suggested that for safety the functions and range should be
selected correctly although protective function for the full measuring
5.When measurement of current, the input current shouldn't be more
than the maximum current labeled on input end.
6. Safety symbols:
risk of high voltage and electric shock!
Contact Person: Rocky